So it became apparent when fiddling with the voltmeter that some slight alteration to my schematics for the power supply would be necessary. The datasheet for the 7107 as well as many posts tell me that it is best to have an isolated supply for the meters. This makes a great deal of sense, this way the meters will be unaffected by whatever happens to the power supply it’s measuring.
My original idea was to tap off the V+, V-, and 0V from the main power supply circuit just after the AC filtering board and just before the Voltage/Current regulator board. Here is the latest schematic with my original idea:
To make an isolated supply, I will have to separate these two using a transformer, meaning it has to be on the AC side. I know what you are thinking (as I thought myself) “oh what a pain, I need another transformer/rectifier/filter” but really, it’s not so bad, and in the name of accuracy and better function why not? It’s easy enough to get a transformer, a bridge rectifier and a couple of caps soldered together. Since it’s not used for the main power supply, the current rating can be far less. In addition, I can use it to generate other control voltages I need allowing for a guaranteed clean signal and control of the main power board. So, everybody wins!
I even thought of a way to reduce the number of components and go with a single output transformer so I believe at the end of the day, the benefits and cost will balance nicely. The 7107 requires a dual symmetrical ±5V power supply. In the original schematic above, I use a cascade of positive and negative voltage regulators to achieve this. As it turns out, I happen to have another Intersil part – ICL7660 – which can make a symmetrical supply from a single one no problem. So right off the bat I can leave out three voltage regulators, not bad!
I will have to perform a few more tests to be sure though. I believe I read that the 7660 can sink up to 45mA so I need to monitor the -5V rail on my voltmeter to see if two meters would exceed 80% of this figure. Doubtful since the displays are driven off the +5V supply. Still worth checking though in case I have the current path wrong.
So what I need to do is move the 12 power diodes to the main board (these are there to compensate for the slightly higher voltage from the main transformer and to lower the differential voltage a bit to save heat and preserve linearity of the variable voltage regulator) and instead connect the control voltage board to a smaller, isolated power supply I will make for it. Here’s the laundry list:
- 24V, 1A transformer, single output
- Small rectifier and filter caps to replace the diodes and connect to the transformer
As I mentioned above, this supply is not just for the displays, but also provides two voltage references which I would like to be independent and accurate as these will set the Voltage and Current Limit on the main board. If I get a large drain on the main board, the control voltages will be unaffected. The more I think about this, the more I like it. Also, it will power the 4 relays ensuring proper operation.
I haven’t had time to draw up a new schematic for this scheme, but here’s the flow:
- AC 115-117V transformer gets rectified and filtered to about 26-28Vdc
- LM7824 steps down and regulates this to 24V providing the reference voltage for Vset and the input to the 12V regulator
- LM7812 steps down from 24V and controls all the 12V bits and bobs I have including the 4 relays and powers the soft start and short circuit protection circuits on the output board and provides input for the 5V supply
- LM7805 steps down from 12V powering the +5V rail for the meters and feeds into the ICL7660 for inverting
- ICL7660 inverts the voltage for the -5V rail for the meters
Looks like it may just work. My only concern is the relays providing too low an impedance path to ground which may screw everything up. That’s what testing is for! I’ll post an updated schematic when I have a moment to create one and illustrate this idea a bit better.
Time also to raid my parts, I know i have a bridge rectifier and caps for it and I’m hoping one of my salvaged transformers will do the job. If not, I’ll buy one.